ii Heracell Service Manual – Free download as PDF File .pdf), Text File . txt) or read online for free. incubator heracell SM. HERAcell® CO2 Incubator Operating Instructions. Version: / – 3 -. Contents. 1. Information regarding Safe Operation. Manuals and User Guides for Thermo Scientific Heracell i. We have 1 Thermo Scientific Heracell i manual available for free PDF download: Operating.
|Country:||Turks & Caicos Islands|
|Published (Last):||3 September 2007|
|PDF File Size:||1.68 Mb|
|ePub File Size:||3.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
Thermo Scientific Heracell 240i Operating Instructions Manual
To cancel the auto-start routine: Chapter 14 Device log Page 67 optional Gas connection with gas monitoring system hegacell Figure. To resume the auto-start routine: To toggle between two options : The gas monitoring system remains active even when the CO control is switched off. For a detailed overview of causes for errors and their correction, please refer to the end of this chapter! Page Failure to set the gas-tight screen heracelp in accordance with the actually installed door version may impair the incubation performance.
The system returns to the Configuration menu. Thermo Scientific warrants the operational safety and the operativeness of the device only if installation and repairs are performed properly.
Connect the serial interface cable connector not included in the standard equipment to the socket at the device supply interface at the rear of the device. A pressure compensation opening with insert in the rear panel herafell the device allows a compensation between the pressures in the work space and in the operating room.
Fill the floorpan of the work space with a sufficient quantity of water.
Thermo HERAcell-CO2-Incubator 150i 240i manual en
Changing the user-defined code: When installing the device, make sure that the installation and supply connections are freely accessible. When temperature and relative humidity are constant, the automatic adjustment of the measuring system is performed. The user surface is divided into two main menus: The measuring sample is withdrawn through the sealable measurement opening of the glass door. If the overtemperature protection mechanism is re-activated although the potential fault causes see error table have been eliminated, take the device out of service and contact the technical service center.
Thermo Scientific – i and i CO2 Community, Manuals and Specifications | MedWrench
Locate the installation directory for the program. Page 99 Changing the user-defined code: Do not spray disinfectant onto the CO sensor in the measuring cell baseplate or onto the O sensor. Place gas humidification immediately to the rear panel of the device and align it parallel to the rear panel. Replacing The Door Seal Z 3 years ago 3 years ago HeraCell Incubator – Fan dead I have 2 of 4 Heracell ‘s purchased in whose fan motors died within 1 month of each other.
The compilation of device informations and the creation of the service file may take some time.
If the STOP key is pressed, the routine is interrupted and the contra-con Stop dialog box is opened as a security hfracell. To indicate the start of this routine, the icon overtemperature is displayed. Page 36 3 additional devices independent of device type CO concentration has been set to 4. Page – Overview of incubator parameters addres The connection of the device to an external alarm system must only be carried out by adequately trained To accept and save the setting: O sensor are integral to the control system of the device.
Instructions of use, instructions for Thermo HERAcell-CO2-Incubator i i manual en
Their measured values are compared to the selected set values. Make sure that the retaining rail taper is positioned correctly in the slot  and that the seal is flush with the door frame.
Room Ventilation When installing the device, make sure that the installation and supply connections are freely accessible. Allow the disinfectant to react as specified by the manufacturer. Unless a user action contact with the pressure-sensitive areas and keys occurs within 30 seconds, the system automatically exits the menu and the most recently confirmed setting defaults.
Page of Go. Note Duration of the procedure: Set value CO 9. Remove Subscription Confirm Remove.
The receptacle for the two device fuses is integral to the socket.